Re: Chronodot 2.1
Post by i_can » 05 Feb 2014 22:19
The question was why the RTC cannot be synchronize?
// following line sets the RTC to the date & time this sketch was compiled
// uncomment it & upload to set the time, date and start run the RTC!
But i still see the serial monitor printing:
since 1970 = 1396827985s = 16166d
now + 7d + 30s: 2014/4/13 23:46:55
and you guessed, that RTC was damaged.
It's possible that the RTC is damaged, then. All I can do at this point is to replace the datalogger. Please email firstname.lastname@example.org
with a link to this thread.
I think its not damaged.
im having the same problem but i dont have the data logger i just have the Chronodot 2.1